The CR200X, our smallest lowest-cost data logger, provides stand-alone operation in harsh, remote environments. It reads input from one or two sensors, then transmits the data using communication peripherals. CRBasic, its full programming language, supports simple or complex programming. Multiple CR200Xs can be configured as a network, or units can be deployed individually.
The CR200X has several input channels for measuring a variety of sensors. Gas discharge tubes provide rugged electrostatic discharge protection for the inputs.
The CR200X does not make differential measurements and is not compatible with SDM devices, multiplexers, or thermocouples (refer to the Compatibility information for compatible sensors, peripherals, and software). Recalibration services are not offered for the CR200X datalogger.
|-NOTE-||Note: Additional specifications are listed in the CR200X-Series Specifications Sheet.|
|Maximum Scan Rate||1 Hz|
|Pulse Count Channels||2|
|Switched Excitation Channels||2 voltage|
|Communication Ports||1 RS-232|
|Switched Battery Port||1|
|Input Voltage Range||0 to 2500 mV|
|Analog Voltage Accuracy||±(0.25% of reading + (1.2 mV)x(offset)) at -40° to +50°C|
|Analog Voltage Resolution||0.6 mV|
|Operating Temperature||-40° to +50°C|
|Power Requirements||7 to 16 Vdc|
|Typical Current Drain||
|Protocols Supported||PakBus (leaf node only), SDI-12|
|CE Compliance Standards to which Conformity Is Declared||IEC61326:2002|
|Dimensions||14.0 x 7.6 x 4.8 cm (5.5 x 3.0 x 1.9 in.)|
|Weight||242 g (8.5 oz)|
|Final Storage||512 kB of flash memory for approximately 125,000 data points|
|Operating System||106 kB of flash memory|
|Intermediate Storage||8 kB of SRAM for communication buffers, calculations, variables, etc.|
Please note: The following shows notable compatibility information. It is not a comprehensive list of all compatible products.
|LOGGERNET||Version 2.1 or higher|
|PC400||Version 1.0 or higher|
|PCONNECT (retired)||Version 3.0 or higher|
|PCONNECTCE (retired)||Version 2.0 or higher|
|VISUALWEATHER||Version 2 or higher|
The CR200X can communicate with a PC via direct connect, NL201 Network Link Interface, NL240 Wi-Fi Network Link, MD485 multidrop modem, and digital cellular modems. Data can be viewed on an iOS device, an Android device, CD295 DataView II Display, or a user-supplied PDA. To use an iOS or Android device, go to the Apple Store or Google Play and download our LoggerLink Mobile Apps free of charge. User-supplied PDAs require either PConnect or PConnectCE software.
The CR200X is not compatible with SDM devices and multiplexers.
The CR200X can measure a variety of sensors including SDI-12 sensors and 4 to 20 mA sensors. It cannot make differential measurements and is not compatible with the freezing-rain/ice, fuel moisture/temperature, geographic position, present weather, soil heat flux, soil matric water potential, and strain sensors listed on our price lists.
Applications with minimal power requirements can use the ENC200 enclosure to house the data logger and the pn 16869 sealed rechargeable battery. The ENC200 cannot house a barometer or a battery that is larger than the pn 16869. However, an ENC10/12 or ENC12/14 enclosure is adequate for most CR200X-based systems.
The CR200X uses an external power supply and has a built-in regulator. When connected to the on-board charging circuit, the sealed rechargeable battery should be 7 A h or smaller. Using larger batteries with the data logger's built-in charger may result in excessive PC board heating. This is especially a concern when the battery is deeply discharged or failing with a shorted cell. Campbell Scientific also recommends that solar panels be 10 W or less and wall chargers be 1 A or smaller.
Execution of this download installs the Operating System and Compiler on your computer for the following dataloggers:CR200X, CR206X, CR211X, CR216X and CR295X. It also updates the support files for the CRBasic Editor.
Note: Newer CR206X dataloggers (serial # ≥ 19122) and newer CR211X dataloggers (serial # ≥ 19143) have 250 mW radios that must use OS 3 or higher for their datalogger operating system.
Number of FAQs related to CR200X: 37
Expand AllCollapse All
The maximum cable length depends on the interface being used.
The voltage excitation channels are high impedance when not connected, meaning that internally they are isolated from the data logger ground. The connected sensor, however, may tie the voltage excitation channel to ground, such as through a completion resistor.
Yes. The simplest method is to use conditional program statements that execute most of the code based on time. For example, the data could be scheduled to log at 6 a.m. and finish at 8 p.m. using CRBasic instructions such as IfTime(). Another option is to use an IfThen/EndIf construction that does a logical test of light-level measurements based on a light sensor. An additional option is to use calculated sunrise and sunset times along with a combination of RealTime() and Case instructions.
For more information, see the “Decisions, Decisions, Decisions…” article.
Not directly. If the CS15-L is connected to a CR200(X) datalogger, the data logger can take the ac measurement and control a solid-state relay based on some threshold within the data logger program. The solid-state relay can then control other relays, solenoids, or motor starters. (Use of a solid-state relay is preferred because the data logger can trigger it with a small 5 Vdc mA signal.) For more information, see the “Measurement and Control Peripherals” section of the operator’s manual.
Yes. Both the RavenXTV (Verizon) and the RavenXTG (AT&T) are available.
They use the same protocol, but the CR800 PakBus protocol is less limited in its capability.
Technically, the SRM-5A is compatible with the CR200X-series dataloggers, but it is more complicated to use it with these data loggers. To use a SRM-5A with a CR200X-series datalogger, the SRM-5A must be in the DTE position on the station modem. In addition, user-supplied adapters and gender changers are needed to complete the connections between the SRM-5A and the RS-232 port of the data logger. Contact Campbell Scientific for more information.
Data will only be lost if the data logger’s internal 3 V battery is also bad. The “table definitions out of date” message simply implies that the data logger is running a program that the PC software does not recognize. After successfully connecting, the message “updating table definitions” should appear. This means that the PC software has knowledge of the program running on the data logger and should not affect data stored on the data logger.
The data logger assigns two attributes for program files:
If the latter option is not set for the data logger, the data logger will not have a program running when it powers up, which will result in lost data. For more information, see the “How Do You Run?” article.
It is possible to upgrade the OS without downloading each OS version between the existing one on the data logger and the most current one available. For example, a CR1000 can be upgraded directly from OS 14 to OS 25. However, when a large jump in OS versions is made, the upgrade will likely restructure the Status table and the Setting table, and they will need to be manually reloaded.
Note that in many instances, an OS update may not be necessary. Before updating, check the OS Revision History to see if a newer version would provide desired benefits.
If small amounts of data are transferred per transmission, it will not be a problem. Larger amounts of data can overrun buffers in the modem, causing lost data. In that situation, lower the baud rate on the data logger to avoid the issue.