Frequently asked questions about our products and solutions
Ask a Question
Ask a Question
(1777 total: showing 1-20)
Why is NIST traceability beneficial for surface temperature sensors, such as the CS240, the 110PV, and the CS240DM?
How quickly can measurements be attained using the ClimaVUE™50?
Does the wind gust measurement interval change for the ClimaVUE™50 based on the query interval?
With the ClimaVUE™50, over what period is wind gust measured?
Is the calibration for the CS240DM NIST traceable?
Are the traditional methods of using RTD to measure the back-of-module temperature not used with the CS240 and CS240DM because these methods may cause hot spots?
Can multiple instances of Campbell Scientific SURVEYOR be run on a PC at the same time?
How tight do the wing-nuts on the CM226 integrated mounting stand need to be tightened to secure the solar sensor?
Can the SG000 determine the exact distance to a lightning strike?
When selecting a recalibration service for a pyranometer, how do the services differ and what is the importance of ISO 9060?
What are some distinguishing features of the MetSENS compact weather sensors?
Can the CSAT3BH heaters be controlled manually?
What happens to the data when the CSAT3BH heaters are on?
How often should the CR1000 be recalibrated?
When should a data acquisition (DAQ) system be used instead of a PC-based system?
How can the age of a sealed rechargeable battery be determined?
If the CS135 reports a warning or alarm, when should a visit to the sensor be made?
Are the CS135 warnings and alarms configurable?
What does a CS135 alarm mean, or how is it defined in the message string?
For the CS135, what does a warning mean, or how is it defined in the message string?