The CS106, manufactured by Vaisala, measures barometric pressure for the range of 500 to 1100 hPa (mBar). This range equates to from below sea level (as in a mine) to over 15,000 feet above sea level. Designed for use in environmental applications, the CS106 is compatible with most Campbell Scientific dataloggers.
The CS106 uses Vaisala’s BAROCAP silicon capacitive sensor to measure barometric pressure. It is encased in a plastic shell (ABS/PC blend) fitted with an intake valve for pressure equilibration.
The CS106 outputs a linear signal of 0 to 2.5 Vdc, which allows the barometer to be directly connected to a Campbell Scientific datalogger. An internal switching circuit allows the datalogger to power the CS106 only during measurement, which reduces power consumption.
|-NOTE-||1 hPa = 1 mBar|
|Pressure Range||500 to 1100 hPa|
|Long-Term Stability||± 0.1 hPa per year|
|Settling Time||1 s to reach full accuracy after power-up|
|Response Time||500 ms to reach full accuracy after a pressure step|
|Output Voltage||0 to 2.5 Vdc|
|Supply Voltage||10 to 30 Vdc|
|Elevation||~609.6 m (2,000 ft) below sea level (as in a mine) to 4,572 m (15,000 ft) above sea level|
|Calibration Uncertainty||±0.15 hPa|
|Temperature Range||-40° to +60°C|
|Dimensions||6.8 x 9.7 x 2.8 cm (2.7 x 3.8 x 1.1 in.)|
|Cable Length||76.2 cm (30 in.)|
|Weight||90 g (3.2 oz)|
Please note: The following shows notable compatibility information. It is not a comprehensive list of all compatible products.
A 2.5-ft cable is supplied with the sensor for connection to the data logger. The cable connects to a control port, single-ended analog input terminal, 12 V power terminal, and ground terminal on the data logger.
The CS106 is typically mounted next to the data logger inside an ENC12/14 or larger enclosure. The very small ENC100 is available for housing the CS106 separately, in its own enclosure.
Number of FAQs related to CS106: 12
Expand AllCollapse All
A CS106 Barometric Pressure Sensor can be added to the station. However, the sensor needs to be wired into the wiring terminal inside the enclosure, and the data logger program needs to be edited to include the instructions to poll the sensor and store the values.
Note: Some versions of the ET107 station that are interfaced with irrigation companies may have proprietary programs or complicated programs that are not easily changed.
The sensor plug on the CS106 is the same as the one on the CS105. If the cable and connections are still working correctly, it should be possible to power down the station and swap out the sensors without any difficulty.
Generally, these sensors are designed to measure the pressure of the environment where they are installed. The CS100 or CS106 can measure the air pressure within a snowpack if the enclosure is adequately sealed away from the snow while allowing the vent to have access to the snow. Note that it may be difficult to accomplish this.
Noisy measurements may be caused by several factors:
Either the CS100 or the CS106 can fit inside an ENC100. For more information, refer to the Specifications information on the ENC100 product page.
In most cases, enclosures and buildings have enough little spaces for the pressure to stay at equilibrium with the outside pressure. Therefore, the sensor does not need to be vented to the outside. It is possible, however, that the rate at which the pressure equalizes is slowed by being in an enclosure, but the rate change usually is not noticeably significant.
If the sensor resides in an absolutely air-tight environment, and the pressure outside this environment needs to be measured, then the sensor will need to be vented to the outside. However, venting to the outside may change the rating of the enclosure or associated inside environment.
The CS106 is not a direct replacement. To use a CS106 in a system that previously used a CS105, the multiplier and offset will need to be changed in the program. Because the CS106 has a longer warm-up time, if the sensor is turned on and off, it may be necessary to make some adjustments to the warm-up time.
The manufacturer recommends calibration on a yearly basis. However, conformance to published standards or regulations may require this on a more frequent basis.
Yes, but only if certain conditions are met: